Clinical Chemistry and Laboratory Medicine (CCLM)
Published in Association with the European Federation of Clinical Chemistry and Laboratory Medicine (EFLM)
Editor-in-Chief: Plebani, Mario
Ed. by Gillery, Philippe / Lackner, Karl J. / Lippi, Giuseppe / Melichar, Bohuslav / Schlattmann, Peter / Tate, Jillian R.
IMPACT FACTOR increased in 2015: 3.017
Rank 5 out of 30 in category Medical Laboratory Technology in the 2014 Thomson Reuters Journal Citation Report/Science Edition
SCImago Journal Rank (SJR) 2015: 0.873
Source Normalized Impact per Paper (SNIP) 2015: 0.982
Impact per Publication (IPP) 2015: 2.238
Evaluation of Commutability of Control Materials
Citation Information: Clinical Chemistry and Laboratory Medicine. Volume 40, Issue 6, Pages 625–630, ISSN (Print) 1434-6621, DOI: 10.1515/CCLM.2002.108, June 2005
- Published Online:
The commutability of 13 control materials was evaluated by performing parallel measurements on two different analysers: a Synchron CX-5 Delta from Beckman-Coulter and a Vitros 950 from Ortho-Clinical Diagnostics. Twenty three clinical chemistry analytes (substrates, electrolytes and enzymatic activities) were determined in plasma from 15 different patients in order to define intermethod relationship for each analyte. The relationship observed for each control material was compared to those obtained for patients' specimens. The results show that commutability depends both on the tested analyte and on the control material. No totally commutable material has been found for the whole set of tested parameters. Most control materials were commutable for inorganic phosphate, glucose, chloride, triglycerides, alanine aminotransferase, amylase and γ-glutamyltransferase, but less than a quarter of control materials were commutable for sodium, calcium, creatinine, alkaline phosphatase and lipase. Seven materials were commutable for more than half of the analytes, whereas five control materials were commutable for less than a quarter of these analytes. We propose to verify the commutability of materials before their use in an external quality control assessement.
Here you can find all Crossref-listed publications in which this article is cited. If you would like to receive automatic email messages as soon as this article is cited in other publications, simply activate the “Citation Alert” on the top of this page.