Clinical Chemistry and Laboratory Medicine (CCLM)
Published in Association with the European Federation of Clinical Chemistry and Laboratory Medicine (EFLM)
Editor-in-Chief: Plebani, Mario
Ed. by Gillery, Philippe / Lackner, Karl J. / Lippi, Giuseppe / Melichar, Bohuslav / Payne, Deborah A. / Schlattmann, Peter / Tate, Jillian R.
IMPACT FACTOR increased in 2015: 3.017
Rank 5 out of 30 in category Medical Laboratory Technology in the 2014 Thomson Reuters Journal Citation Report/Science Edition
SCImago Journal Rank (SJR) 2015: 0.873
Source Normalized Impact per Paper (SNIP) 2015: 0.982
Impact per Publication (IPP) 2015: 2.238
Analytical performance of a new two-step ADVIA Centaur® estradiol immunoassay during ovarian stimulation
Citation Information: Clinical Chemical Laboratory Medicine. Volume 44, Issue 1, Pages 105–109, ISSN (Online) 1437-4331, ISSN (Print) 1434-6621, DOI: 10.1515/CCLM.2006.020, January 2006
- August 18, 2005
- October 18, 2005
Measurement of estradiol is useful in women undergoing ovarian stimulation for in vitro fertilization and embryo transfer (IVF-ET). The analytical performance of a new two-step estradiol assay (ADVIA Centaur® estradiol-6 III from Bayer Diagnostics) was evaluated in 41 sera from 11 women undergoing ovarian stimulation. The results were compared to those obtained with two radioimmunoassays (RIAs; RIA Estradiol Immunotech IM 1663 from Beckmann Coulter and Coat-A-Count Estradiol from Diagnostic Products Corporation) and with one chemiluminescent immunoassay (CLIA; ADVIA Centaur® estradiol-6). The ADVIA Centaur® estradiol-6 III assay was the most sensitive assay, with a functional sensitivity of 55 pmol/L. Within- and between-run coefficients of variation calculated for the new ADVIA Centaur® assay ranged from 3.3% to 9%, which was better than the precision obtained for the other assays. A dilution test showed serum interferences when estradiol was measured in non-diluted samples. No statistical difference was observed between the estradiol results obtained in diluted sera with the new two-step ADVIA Centaur® assay and those measured with the Immunotech RIA and the other CLIA. In conclusion, this new, two-step estradiol assay performed on the ADVIA Centaur® system displays suitable sensitivity, precision and intermethod agreement with the Immunotech RIA for the measurement of serum estradiol concentrations in women undergoing ovarian stimulation and IVF-ET. For correct linearity, estradiol measurement should be performed on diluted samples.