Clinical Chemistry and Laboratory Medicine (CCLM)
Published in Association with the European Federation of Clinical Chemistry and Laboratory Medicine (EFLM)
Editor-in-Chief: Plebani, Mario
Ed. by Gillery, Philippe / Lackner, Karl J. / Lippi, Giuseppe / Melichar, Bohuslav / Schlattmann, Peter / Tate, Jillian R.
IMPACT FACTOR increased in 2015: 3.017
Rank 5 out of 30 in category Medical Laboratory Technology in the 2014 Thomson Reuters Journal Citation Report/Science Edition
SCImago Journal Rank (SJR) 2015: 0.873
Source Normalized Impact per Paper (SNIP) 2015: 0.982
Impact per Publication (IPP) 2015: 2.238
Immunochemical quantification of free immunoglobulin light chains from an analytical perspective
Citation Information: Clinical Chemical Laboratory Medicine. Volume 44, Issue 5, Pages 522–532, ISSN (Online) 1437-4331, ISSN (Print) 1434-6621, DOI: 10.1515/CCLM.2006.118, May 2006
- January 14, 2006
- March 15, 2006
Immunoglobulin light chains are components of antibodies, but some exist in a free form in serum and urine as a result of their excess production over heavy chains. Free light chain (FLC) levels are of the order of milligram per liter in normal serum and urine, but marked increases have been observed in various disease conditions. It has now been established that the measurement of FLC levels contributes to diagnosis and clinical management in monoclonal gammopathies. Recent developments in FLC assays have been adapted to several automated platforms and they have now become available in laboratories. There have, however, been some concerns regarding the analytical aspects. The current assay specificity appears to be insufficient to prevent the influence of intact light chains of several orders of magnitude greater than FLCs in serum. Moreover, the heterogeneous nature of light chains makes accurate quantification unreliable. FLC assays have never been standardized because of the lack of an international reference calibrator. In this review, we summarize the reports on FLC measurements and examine the specificity of anti-FLC antibodies and the reliability of FLC assays. We also discuss difficulties in the standardization and setting of normal reference intervals for FLC assays.
Here you can find all Crossref-listed publications in which this article is cited. If you would like to receive automatic email messages as soon as this article is cited in other publications, simply activate the “Citation Alert” on the top of this page.