Clinical Chemistry and Laboratory Medicine (CCLM)
Published in Association with the European Federation of Clinical Chemistry and Laboratory Medicine (EFLM)
Editor-in-Chief: Plebani, Mario
Ed. by Gillery, Philippe / Lackner, Karl J. / Lippi, Giuseppe / Melichar, Bohuslav / Payne, Deborah A. / Schlattmann, Peter / Tate, Jillian R.
IMPACT FACTOR increased in 2015: 3.017
Rank 5 out of 30 in category Medical Laboratory Technology in the 2014 Thomson Reuters Journal Citation Report/Science Edition
SCImago Journal Rank (SJR) 2015: 0.873
Source Normalized Impact per Paper (SNIP) 2015: 0.982
Impact per Publication (IPP) 2015: 2.238
Sensitivity and specificity of the high fluorescent lymphocyte count-gate on the Sysmex XE-5000 hematology analyzer for detection of peripheral plasma cells
1Clinical Laboratory, Sint Lucas Andreas Hospital, Amsterdam, The Netherlands
Citation Information: Clinical Chemistry and Laboratory Medicine. Volume 49, Issue 4, Pages 685–688, ISSN (Online) 1437-4331, ISSN (Print) 1434-6621, DOI: 10.1515/CCLM.2011.100, March 2011
- Published Online:
Background: Plasma cells are one of the end products of the B-lymphocyte mediated immune response. These cells normally reside in the bone marrow or some peripheral lymphoid tissues. Increased numbers of plasma cells in the blood, usually indicates pathology such as infection, auto-immunity or haematological malignancy. Therefore, the ability to measure plasma cells (PCs) on an automated cell analyzer might be advantageous.
Methods: The performance of the Sysmex XE-5000 leukocyte differential channel [high fluorescent lymphocyte count (HFLC) area] was evaluated for the ability to detect plasma cells in peripheral blood and compared to the detection of plasma cells by flow cytometric analyses.
Results: Our results show that the HFL count from the XE-5000 correlates (R2=0.8) with the number of PCs in peripheral blood, but detects PCs with moderate to good sensitivity (88.9%) and specificity (87.8%).
Conclusions: The Sysmex XE-5000 is suitable for screening blood samples for the presence of elevated number of plasma cells in peripheral blood, but the actual quantification needs to be confirmed by flow cytometry.
Here you can find all Crossref-listed publications in which this article is cited. If you would like to receive automatic email messages as soon as this article is cited in other publications, simply activate the “Citation Alert” on the top of this page.