Clinical Chemistry and Laboratory Medicine (CCLM)
Published in Association with the European Federation of Clinical Chemistry and Laboratory Medicine (EFLM)
Editor-in-Chief: Plebani, Mario
Ed. by Gillery, Philippe / Lackner, Karl J. / Lippi, Giuseppe / Melichar, Bohuslav / Schlattmann, Peter / Tate, Jillian R.
IMPACT FACTOR increased in 2015: 3.017
Rank 5 out of 30 in category Medical Laboratory Technology in the 2014 Thomson Reuters Journal Citation Report/Science Edition
SCImago Journal Rank (SJR) 2015: 0.873
Source Normalized Impact per Paper (SNIP) 2015: 0.982
Impact per Publication (IPP) 2015: 2.238
Lupus anticoagulant testing: analyzing fresh samples after a single centrifugation and after a 6–8 h delay
1Regional Laboratory, Haifa and the Western Galilee, Clalit Health Services, Nesher, Israel
Citation Information: Clinical Chemistry and Laboratory Medicine (CCLM). Volume 50, Issue 2, Pages 367–370, ISSN (Online) 1437-4331, ISSN (Print) 1434-6621, DOI: 10.1515/cclm.2011.768, October 2011
- Published Online:
Background: For automation it is important to know the effect of analyzing fresh samples after a single rather than after two centrifugations, and to determine test stability.
Methods: This study compared silica clotting times (SCT) and dilute Russel’s viper venom tests (dRVVT) after one and two centrifugations in 50 fresh plasma samples. Then it compared test results within 4 h to those after 6–8 h of blood drawing in 40 samples. Means, minimums, maximums and quartiles of the paired screen were compared, test ratios were confirmed and correlations, linear regressions and Bland-Altman statistics were calculated.
Results: The distributions of test results were nearly identical, regardless of the number of centrifugations or timing of the analysis. The first centrifugation explained 97.7% and 94.8% of the variance of tests results after the second centrifugation for the SCT and dRVVT, respectively. The test results after 6–8 h explained 98.3% and 96.3% of the variance of the SCT-ratios and dRVVT-ratios, respectively, tested within 4 h. Inter-day coefficients of correlation of ratio comparisons were similar to those of the controls values.
Conclusions: Testing of fresh samples after a single centrifugation might replace batch testing of frozen samples after double centrifugation, providing timelier reporting of results and resulting in savings of technician time.