Titania layers obtained by a sol-gel technique doped with redox mediator, Meldola’s Blue, were employed for construction of a new NADH senor. Optimization of preparation process as well as experimental conditions affecting the response of the sensor were examined. Under optimal conditions NADH could be determined in the wide linear range from 90 to 2300 µM with detection limit 12 µM and a high sensitivity 12.5 nA µM−1. The usefulness of developed sensor was preliminarily checked in determination of NADH forming during enzymatic oxidation of ethanol catalyzed by alcohol dehydrogenase (ADH).
 T. Tsujimoto, T. Yabutani, A. Sano, Y. Tani, H. Murotani, Y. Mishima, K. Maruyama, M. Yasuzawa, J. Motonaka, Anal. Sci. 23, 59 (2007) http://dx.doi.org/10.2116/analsci.23.5910.2116/analsci.23.59Search in Google Scholar
 C.M. Maroneze, L.T. Arenas, R.S.C. Luz, E.V. Benvenutti, R. Landers, Y. Gushikem, Electrochim. Acta 53, 4167 (2008) http://dx.doi.org/10.1016/j.electacta.2007.12.07210.1016/j.electacta.2007.12.072Search in Google Scholar
 A.S. Santos, L. Gorton, L.T. Kubota, Electroanalysis 14, 805 (2002) http://dx.doi.org/10.1002/1521-4109(200206)14:12<805::AID-ELAN805>3.0.CO;2-P10.1002/1521-4109(200206)14:12<805::AID-ELAN805>3.0.CO;2-PSearch in Google Scholar
 A. Vasilescu, S. Andreescu, C. Bala, C.S. Litescu, T. Noguer, J.-L. Marty, Biosens. Bioelectron. 18, 781 (2003) http://dx.doi.org/10.1016/S0956-5663(03)00044-710.1016/S0956-5663(03)00044-7Search in Google Scholar
© 2011 Versita Warsaw
This work is licensed under the Creative Commons Attribution-NonCommercial-NoDerivatives 3.0 License.