Clinical Chemistry and Laboratory Medicine (CCLM)
Published in Association with the European Federation of Clinical Chemistry and Laboratory Medicine (EFLM)
Editor-in-Chief: Plebani, Mario
Ed. by Gillery, Philippe / Lackner, Karl J. / Lippi, Giuseppe / Melichar, Bohuslav / Payne, Deborah A. / Schlattmann, Peter / Tate, Jillian R.
12 Issues per year
IMPACT FACTOR increased in 2015: 3.017
Rank 5 out of 30 in category Medical Laboratory Technology in the 2014 Thomson Reuters Journal Citation Report/Science Edition
SCImago Journal Rank (SJR) 2015: 0.873
Source Normalized Impact per Paper (SNIP) 2015: 0.982
Impact per Publication (IPP) 2015: 2.238
Comparison of the Abbott IMx® and a High-Performance Liquid Chromatography Method for Measuring Total Plasma Homocysteine
The determination of the total concentration of plasma homocysteine is of interest in a variety of clinical circumstances, especially, in the evaluation of the risk of cardiovascular disease. However, most of the methods available to date, many of them chromatographic, are not well suited for the majority of clinical laboratories. Several automated methods are now or will be, shortly, commercially available. We have compared one of them, the fluorescence polarization immunoassay (FPIA) adapted to the IMx® analyzer (Abbott Laboratories), with the high-performance liquid chromatography (HPLC) method with fluorescent detection currently used in our laboratory. The results show that the FPIA-IMx® method is less imprecise and slightly more sensitive than the HPLC. The comparison of 67 clinical plasma specimens indicated that there is a proportional error disagreement between FPIA-IMx® and HPLC (FPIA=1.19 HPLC + 0.92; confidence region for slope and y-intercept were, respectively, from 1.06 to 1.31 and from−0.06 to 2.32). The nature of this error is not explained by the experiments performed to study the inaccuracy of both methods, which included the investigation of dilution parallelism, analytical recovery and cross-reactivity. The different results of homocysteine concentration obtained with FPIA-IMx® and HPLC must be taken into account when a change of methodology is under consideration.
Here you can find all Crossref-listed publications in which this article is cited. If you would like to receive automatic email messages as soon as this article is cited in other publications, simply activate the “Citation Alert” on the top of this page.