Clinical Chemistry and Laboratory Medicine (CCLM)
Published in Association with the European Federation of Clinical Chemistry and Laboratory Medicine (EFLM)
Editor-in-Chief: Plebani, Mario
Ed. by Gillery, Philippe / Lackner, Karl J. / Lippi, Giuseppe / Melichar, Bohuslav / Payne, Deborah A. / Schlattmann, Peter / Tate, Jillian R.
IMPACT FACTOR increased in 2015: 3.017
Rank 5 out of 30 in category Medical Laboratory Technology in the 2014 Thomson Reuters Journal Citation Report/Science Edition
SCImago Journal Rank (SJR) 2015: 0.873
Source Normalized Impact per Paper (SNIP) 2015: 0.982
Impact per Publication (IPP) 2015: 2.238
Measurement of serum and plasma osmolality in healthy young humans – influence of time and storage conditions
Citation Information: Clinical Chemistry and Laboratory Medicine. Volume 42, Issue 8, Pages 927–932, ISSN (Print) 1434-6621, DOI: https://doi.org/10.1515/CCLM.2004.150, June 2005
- February 23, 2004
- June 15, 2004
- Published Online:
Background: The precise measurement of osmolality is crucial in the differential diagnosis of disorders of water balance. Storage conditions, and freezing and thawing of serum or plasma samples before osmometry may influence the accuracy of measured values. Methods: A series of serum and plasma samples of 25 healthy young individuals were stored under different conditions at different temperatures (room temperature (22°C), 7°C, –21°C, –78°C) for up to 56 days. Before freezing a protein-stabilizing agent (bacitracin) was added to one part of the samples. Osmolality was examined using the freezing point method. Results: At room temperature osmolality was stable for up to 3 days but showed a tendency toward an increase that was significant on day 14. In contrast, at 7°C an initial significant decrease in serum osmolality occurred (day 1), which was followed by a slow increase. Serum samples stored at –21°C showed a significantly lower osmolality on the 14th day compared to baseline. Adding bacitracin before freezing reduced this decrease by more than half, but the deviation was still significant. In samples stored at –78°C no significant alteration of osmolality from baseline was observed over the observation period of 56 days if samples were thawed in a 37°C water bath. Conclusion: Immediate measurement of osmolality is most reliable in order to obtain accurate values, although storing at room temperature does not influence osmolality significantly during the first 3 days. If storage is necessary for longer, samples should be stored at –78°C and must be thawed quickly (at 37°C). Under these conditions reliable values can be obtained from frozen serum or plasma. Storage at 7°C is not recommended. If samples are stored at –21°C the addition of a protein-stabilizing agent may be useful.
Here you can find all Crossref-listed publications in which this article is cited. If you would like to receive automatic email messages as soon as this article is cited in other publications, simply activate the “Citation Alert” on the top of this page.