Clinical Chemistry and Laboratory Medicine (CCLM)
Published in Association with the European Federation of Clinical Chemistry and Laboratory Medicine (EFLM)
Editor-in-Chief: Plebani, Mario
Ed. by Gillery, Philippe / Lackner, Karl J. / Lippi, Giuseppe / Melichar, Bohuslav / Payne, Deborah A. / Schlattmann, Peter / Tate, Jillian R.
IMPACT FACTOR increased in 2015: 3.017
Rank 5 out of 30 in category Medical Laboratory Technology in the 2014 Thomson Reuters Journal Citation Report/Science Edition
SCImago Journal Rank (SJR) 2015: 0.873
Source Normalized Impact per Paper (SNIP) 2015: 0.982
Impact per Publication (IPP) 2015: 2.238
Automated analysis of pleural fluid total and differential leukocyte counts with the Sysmex XE-2100
Citation Information: Clinical Chemical Laboratory Medicine. Volume 44, Issue 11, Pages 1367–1371, ISSN (Online) 1437-4331, ISSN (Print) 1434-6621, DOI: 10.1515/CCLM.2006.242, November 2006
- May 11, 2006
- August 29, 2006
- Published Online:
Background: Determination of leukocyte (WBC) counts in pleural fluid is routinely performed by microscopic examination. In this study, we evaluated the performance of automated (differential) WBC counting in comparison with manual counting.
Methods: Pleural fluid samples (n=45) were obtained from patients undergoing diagnostic thoracocentesis. The manual total WBC count was determined after Samson staining in a Fuchs-Rosenthal hemocytometer; microscopic differential counts were performed on May-Grünwald Giemsa-stained cytospin slides. The Sysmex XE-2100 hematology analyzer was used for automated (differential) WBC counting. The functional detection limit was determined by serial dilution of continuous ambulatory peritoneal dialysis (CAPD) fluid and replicate measurements of each dilution.
Results: The automated WBC count (×106/L) was highly correlated with that of the microscopic reference method (r2=0.95; WBC-analyzer=0.97×WBC-reference method+16; n=45). Good agreement was also observed for the absolute lymphocyte count (r2=0.92; WBC-analyzer=0.99×WBC-reference method+32; n=36), neutrophil count (r2=0.94; WBC-analyzer=0.91×WBC-reference method+6; n=35), and monocyte count (r2=0.73; WBC-analyzer=0.83×WBC-reference method+6; n=38). The functional detection limit for WBCs was calculated at 50×106/L (coefficient of variation 20%).
Conclusions: With some limitations, total and differential WBC counts in pleural fluid can be reliably determined using the Sysmex XE-2100 instrument.
Clin Chem Lab Med 2006;44:1367–71.
Here you can find all Crossref-listed publications in which this article is cited. If you would like to receive automatic email messages as soon as this article is cited in other publications, simply activate the “Citation Alert” on the top of this page.