Clinical Chemistry and Laboratory Medicine (CCLM)
Published in Association with the European Federation of Clinical Chemistry and Laboratory Medicine (EFLM)
Editor-in-Chief: Plebani, Mario
Ed. by Gillery, Philippe / Lackner, Karl J. / Lippi, Giuseppe / Melichar, Bohuslav / Payne, Deborah A. / Schlattmann, Peter / Tate, Jillian R.
IMPACT FACTOR increased in 2015: 3.017
Rank 5 out of 30 in category Medical Laboratory Technology in the 2014 Thomson Reuters Journal Citation Report/Science Edition
SCImago Journal Rank (SJR) 2015: 0.873
Source Normalized Impact per Paper (SNIP) 2015: 0.982
Impact per Publication (IPP) 2015: 2.238
Validation of the body fluid module on the new Sysmex XN-1000 for counting blood cells in cerebrospinal fluid and other body fluids
1Department of Clinical Chemistry, Erasmus MC, University Medical Center Rotterdam, Rotterdam, The Netherlands
Citation Information: Clinical Chemistry and Laboratory Medicine. Volume 50, Issue 10, Pages 1791–1798, ISSN (Online) 1437-4331, ISSN (Print) 1434-6621, DOI: 10.1515/cclm-2011-0927, May 2012
- Published Online:
Background: We evaluated the body fluid (BF) module on the new Sysmex XN-1000 for counting blood cells.
Methods: One hundred and eighty-seven BF samples [73 cerebrospinal fluid (CSF), 48 continuous ambulatory peritoneal dialysis (CAPD), 46 ascites, and 20 pleural fluid] were used for method comparison between the XN-1000 and manual microscopy (Fuchs-Rosenthal chamber and stained cytospin slides) for counting red blood cells (RBCs) and white blood cells (WBCs) (differential).
Results: Good agreement was found for counting WBCs (y=1.06x+0.09, n=67, R2=0.96) and mononuclear cells (MNs) (y=1.04x–0.01, n=40, R2=0.93) in CSF. However, the XN-1000 systematically counted more polymorphonuclear cells (PMNs) (y=1.48x+0.18, n=40, R2=0.99) compared to manual microscopy. Excellent correlation for RBCs >1×109/L (y=0.99x+116.56, n=26, R2=0.99) in CSF was found. For other fluids (CAPD, ascites and pleural fluid) excellent agreement was found for counting WBCs (y=1.06x+0.26, n=109, R2=0.98), MNs (y=1.06x–0.41, n=93, R2=0.96), PMNs (y=1.06x+0.81, n=93, R2=0.98) and RBCs (y=1.04x+110.04, n=43, R2=0.98). By using BF XN-check, the lower limit of quantitation (LLoQ) for WBC was defined at 5×106/L. Linearity was excellent for both the WBCs (R2=0.99) and RBCs (R2=0.99) and carry-over never exceeded 0.05%.
Conclusions: The BF module on the XN-1000 is a suitable tool for fast and accurate quantification of WBC (differential) and RBC counts in CSF and other BFs in a diagnostic setting.
Here you can find all Crossref-listed publications in which this article is cited. If you would like to receive automatic email messages as soon as this article is cited in other publications, simply activate the “Citation Alert” on the top of this page.