Briggs CJ, Linssen J, Longair I, Machin SJ. Improved flagging rates on the Sysmex XE-5000 compared with the XE-2100 reduce the number of manual film reviews and increase laboratory productivity. Am J Clin Pathol 2011;136:309–16. [Crossref] [Web of Science]
Hilborne LH, Wenger NS, Oye RK. Physician performance of laboratory tests in self-service facilities. Residents’ perceptions and performance. J Am Med Assoc 1990;264:382–6.
Cornet E, Perol JP, Troussard X. Performance evaluation and relevance of the CellaVision DM96 system in routine analysis and in patients with malignant hematological diseases. Int J Lab Hematol 2008;30:536–42. [Crossref] [Web of Science]
Rumke CL. Imprecision of ratio-derived differential leukocyte counts. Blood Cells 1985;11:311–5. [PubMed]
Perel ID, Herrmann NR, Watson LJ. Automated differential leucocyte counting by the Geometric Data Hematrak system: eighteen months experience in a private pathology laboratory. Pathology 1980;12:449–60. [Crossref]
Ceelie H, Dinkelaar RB, van Gelder W. Examination of peripheral blood films using automated microscopy; evaluation of Diffmaster Octavia and Cellavision DM96. J Clin Pathol 2007;60:72–9. [Crossref] [Web of Science]
Kratz A, Bengtsson HI, Casey JE, Keefe JM, Beatrice GH, Grzybek DY, et al. Performance evaluation of the CellaVision DM96 system: WBC differentials by automated digital image analysis supported by an artificial neural network. Am J Clin Pathol 2005;124:770–81. [Crossref]
Yamamoto T, Tabe Y, Ishii K, Itoh S, Maeno I, Matsumoto K, et al. Performance evaluation of the CellaVision DM96 system in WBC differentials. Rinsho Byori 2010;58:884–90 [in Japanese].
Satoh M, Ishii K, Tabe Y, Yamamoto T, Horii T, Miida T, et al. Influence of the smudge cells to the WBC differential count after storage in a collection tube [Japanese]. J Jap Soc Lab Hematol 2012;13:304–10.
Clinical and Laboratory Standards Institute (CLSI). Reference leukocytes (WBC) differential count (proportional) and evaluation of instrumental methods: approved standard, 2nd ed. CLSI Document H20-A2. Wayne, PA: CLSI, 2007.
Briggs C, Longair I, Slavik M, Thwaite K, Mills R, Thavaraja V, et al. Can automated blood film analysis replace the manual differential? An evaluation of the CellaVision DM96 automated image analysis system. Int J Lab Hematol 2009;31:48–60. [Crossref]
Lin MY, Weinstein RA, Hota B. Delay of active antimicrobial therapy and mortality among patients with bacteremia: impact of severe neutropenia. Antimicrob Agents Chemother 2008;52:3188–94. [Web of Science]
Freifeld AG, Bow EJ, Sepkowitz KA, Boeckh MJ, Ito JI, Mullen CA, Raad II, et al. Clinical practice guideline for the use of antimicrobial agents in neutropenic patients with cancer: 2010 update by the Infectious Diseases Society of America. Clin Infect Dis 2011;52:e56–93. [Crossref]
Simson E, Gascon-Lema MG, Brown DL. Performance of automated slidemakers and stainers in a working laboratory environment – routine operation and quality control. Int J Lab Hematol 2009;32:e64–76. [Web of Science] [Crossref]
Honnma H, Ishii K, Tabe K, Kusama I, Maeno I, Horii T, et al. Evaluation of Basic Performance and Red Cell Size Information of Sysmex XE5000. J Jap Soc Lab Hematol 2009;10:367–74 [in Japanese].
Hur M, Cho JH, Kim H, Hong MH, Moon HW, Yun YM, et al. Optimization of laboratory workflow in clinical hematology laboratory with reduced manual slide review: comparison between Sysmex XE-2100 and ABX Pentra DX120. Int J Lab Hematol 2011;33:434–40.
Bitencourt ED, Voegeli CF, Onzi GS, Boscato SC, Ghem C, Munhoz T. Validation of the Sysmex sp-1000i automated slide preparer-stainer in a clinical laboratory. Rev Bras Hematol Hemoter 2013;35:404–8. [Crossref]